MDF® 808 Professional Aneroid Sphygmomanometer with Stethoscope
MDF® 808M Calibra Aneroid Sphygmomanometer
MDF® 808B Calibra Pro Professional Aneroid Sphygmomanometer > Double Bellow
MDF® 830 Mobile Aneroid Sphygmomanometer
MDF® 840 Desk & Wall Aneroid Sphygmomanometer
MDF® 848XP Medic PalmT Aneroid Sphygmomanometer
Read instructions carefully before using and save for future reference.
General Safety: MDF® sphygmomanometers are designed exclusively as a non-invasive method of measuring blood pressure on healthy skin of the upper arm or thigh regions only.
Do not inflate past 300 mmHg Always consult a physician
Medical professional use only Do not leave cuff inflated on arm or thigh
Limit use to less than 2 minutes Do not leave unattended around children
General Usage: Prior to each use, make sure all parts are securely fastened and properly connected. Always use with a stethoscope for accurate and complete blood pressure measurement (stethoscope is included with MDF® 808 only ).
Cuff Placement: Place cuff midway between shoulder and elbow with a gap of 2 to 3 cm above the elbow. Locate the brachial artery pulse with first two fingers and place cuff so the arrow of the arterial indicator is directly over the brachial artery. The cuff should be at the same height as the heart.
Cuff Inflation: Screw valve clockwise to shut air release valve. lnflate cuff to 20-30 mm above expected systolic pressure.
Stethoscope Placement: Place the eartips of stethoscope into ears. The headset should be worn with the eartips pointed away from you if it is held in front of you. Place the chestpiece of the stethoscope firmly and directly over brachial artery underneath the cuff. If the pulse is heard prior to releasing the valve, inflate the cuff another 20 mmHg or until no pulse is heard.
Measurement: Screw valve counter-clockwise to release valve at a rate of 2-3 mmHg per second. The systolic pressure is the gauge reading at the first audible pulse. Continue with air release. The diastolic pressure is the gauge reading at the last audible pulse.
General Cleaning & Care: Clean soiled parts of sphygmomanometer thoroughly before each use. All parts of the sphygmomanometer and stethoscope can be cleaned with a clean cotton cloth and isopropyl alcohol solution or water with mild detergent. Do not let any liquid enter the inside of sphygmomanometer. Cuff cover can be hand washed with warm water and mild detergent. Air dry; do not machine dry.
Never inflate past 300 mmHg
Keep sharp objects away from cuff
Never disassemble the sphygmomanometer
Avoid impact or shock to the sphygmomanometer
Always deflate cuff completely after measurement and prior to storage
Avoid prolong exposure to excessive heat, cold, solvents, moisture, and oils
The pointer of a calibrated gauge will rest on the zero at no pressure
If pointer does not rest on zero, please contact the manufacturer at MDFinstruments.com for an authorized service center near you
Frequently check for and remove dust from screen filter of intake valve at end of inflation bulb
The sphygmomanometer should be stored between -20 °C and 70 °C -20℃ humidity of 85%
The sphygmomanometer should be used at temperatures between 10 °C and 40 °C at a relative humidity of 85%
Our commitment to high-quality instruments is backed by our lifetime warranty on stethoscopes.
Range: 0-300 mmHg
Accuracy: +/- 3 mmH
This device complies with EN 1060-1:1995+A1:2002 & EN 1060-2:1995+A1:2009
Your device should be checked for proper calibration and performance every two years.Please contact Authorized Representative in European Community below for the service center near you. Do not attempt to calibrate or repair this device yourself.
The sphygmomanometer should be calibrated at temperature range of 15 °C to 25 °C and a relative humidity range of 20% to 85%.
The sphygmomanometer should not leak air in excess of 4 mmHg/min (0.5kPa/min) when the valve is closed.
The maximum error of the measurement of the cuff pressure at any point of the scale range should be within ±3 mmHg (±0.4 kPa).